Re: Re : Re: accelerometer calibration
W. B. Kranendonk
wankelwankel at yahoo.com
Sat Dec 25 22:18:51 CET 2010
--- On Sat, 12/25/10, Timo Juhani Lindfors <timo.lindfors at iki.fi> wrote:
> "W. B. Kranendonk" <wankelwankel at yahoo.com>
> > Would that mean that (one of) the meter(s) is
> connected at an angle?
> > Are they not calibrated in software?
> top and bottom accelerometers are oriented differently
Yes, now you say it I recall reading about it. I actually meant: could it be that it is installed at an incorrect angle, i.e. not perpendicular or paralell to the case, but slanted.
> > Is the accelerometer-dump standalone, and do you have
> a binary version available?
I found it after searching for accelerometer-dump, but (the FR) didn't have gcc installed and don't know yet which packages to install to make it useful. At least a reason now to dive into figuring that out :-)
Any idea what to do about the diversion? I bought that FR second hand, and it has had this "feature" ever since I got it.
More information about the community